Thick paint putty consistency measurement is the best putty detection instrument for determining the thin consistency of thick paint or putty,which is simple,intuitive and accurate.
Thick paint putty consistency tester structure and working principle:.
Take a quantitative volume(20ml)of the Rong Jida instrument sample and place it between two glass plates.With a certain weight pressure,after a certain period of time,it can be expressed according to the degree of spreading of the sample and the diameter of the diffusion.Its consistency,if the degree of spread is poor,and the diffusion diameter is small,it means that it has a high consistency;otherwise,it means that its consistency is low,and it is an effective identification tool for comparing the thin consistency of different samples.
QCT thick paint putty consistency tester specifications and use requirements:
1.The inner diameter of the drum is 29.7mm,the depth is 29mm,and the drum volume is guaranteed to be 20.07ml.
2.The upper glass pressing plate is 200×200mm,the thickness is 5mm,and a circle with a diameter of 5-16cm is engraved on the plate.
3.The weight is 2000g.
4.The operating environment temperature is 25±1℃,and the relative humidity is 60%-70%.
5.Lacquer knife and stopwatch for test(provided).